Emerson CON011 + PR6424/01 | Eddy Current Signal Converter

2026-04-10T10:21:00+08:00

Vendor: Epro
Type: Eddy Current Technology
Price: 350.00

Product Details

Product Overview

The Emerson Epro CON011 + PR6424/01 kit delivers a robust, non-contacting solution for critical machinery protection. This system combines the PR6424/01 8mm eddy current proximity probe with the CON011 signal converter. It measures shaft vibration, axial displacement, and rotational position by detecting microscopic changes in the gap between the probe tip and the target material. The CON011 converter then transforms these high-frequency oscillations into industry-standard analog signals. This setup serves as the primary data acquisition layer for distributed control systems (DCS) and predictive maintenance platforms in power plants and refineries.

Technical Specifications

Attribute | Value
Manufacturer | Emerson (Epro Brand)
Model Number | CON011 + PR6424/01
Sensor Type | Eddy current proximity probe (8mm)
Measurement Range | Typically 0–2 mm displacement
Frequency Response | DC to 10 kHz
Output Signal | 4–20 mA or standardized voltage
Cable Length | 5 m (Integrated with PR6424/01)
Power Supply | 24 VDC (via CON011)
Probe Temperature | –35°C to +180°C
Converter Temperature | 0°C to +70°C
Dimensions | 3cm * 19cm * 12.8cm (Module)
Weight | 0.3 kg (Net)
Condition | 100% Brand New Original

Engineering Advantages

High-Fidelity Signal Processing: The CON011 converter utilizes advanced galvanic isolation to separate the sensor input from the system output. This architecture blocks electrical noise and ground loops, ensuring that the monitoring system receives accurate waveforms even in high-EMI industrial environments.

Non-Contact Wear-Free Monitoring: The eddy current technology detects shaft movement without physical contact. This eliminates mechanical wear on the probe and target, allowing for permanent, maintenance-free monitoring of high-speed rotors.

Thermal Endurance: The PR6424/01 probe tip withstands extreme temperatures up to +180°C. This capability allows for installation in the immediate vicinity of turbine bearings and steam-driven components where standard electronics fail.

Compact DIN-Rail Deployment: The CON011 module features a slim, 3cm width profile. This compact design allows engineers to mount multiple converters in narrow local junction boxes or high-density control cabinets without upgrading rack space.

Explosive Atmosphere Safety: ATEX and CE certifications ensure that this sensor system meets the stringent safety requirements for hazardous areas in the oil, gas, and petrochemical sectors.

FAQs

Q1: Is the Emerson Epro CON011 + PR6424/01 kit brand new?

A1: Yes. We supply only 100% brand new, original factory units. Every kit arrives in original packaging and includes a full 12-month industrial warranty.

Q2: What is the primary difference between the CON010 and the CON011?

A2: While both serve as Epro signal converters, the CON011 often features updated internal circuitry or specific calibration profiles optimized for newer generations of the PR642x sensor series. Always check your specific system manual to ensure the converter matches your I/O card requirements.

Q3: Can I cut the 5m integrated cable if it is too long?

A3: No. You must never cut or shorten the proximity probe cable. The length is a critical part of the sensor's tuned circuit; altering the cable length will invalidate the calibration and result in inaccurate displacement readings.

Q4: Does this system support Modbus or digital communication?

A4: The CON011 provides a standardized analog output (4-20 mA or voltage). To get digital data like Modbus, you must connect the CON011 output to a PLC or a machinery health monitor like the Emerson AMS 6500.