CON010 + PR6424/010-040 | Shaft Vibration & Displacement Kit

2026-04-10T10:15:05+08:00

Vendor: Epro
Type: Vibration Monitoring Systems
Price: 350.00

Product Details

Product Overview

The Emerson Epro CON010 + PR6424/010-040 system delivers a high-precision, non-contact solution for monitoring shaft vibration, axial displacement, and rotational position in critical machinery. This industrial-grade kit pairs the PR6424/010-040 eddy current proximity probe with the CON010 signal converter module. The sensor utilizes electromagnetic fields to detect microscopic changes in the distance between the probe tip and the target shaft. The CON010 module then processes these fluctuations, outputting standardized signals to machinery protection systems or plant-wide DCS platforms. It serves as a fundamental component for the predictive maintenance of high-speed turbines, compressors, and pumps.

Part Number Breakdown: PR6424/010-040

PR6424: 8mm Eddy Current Proximity Sensor series for non-contacting measurement.

/010: Designates the specific cable and thread configuration (Integrated 5m cable).

-040: Indicates the sleeve/housing length (40mm) for precise mounting depth adjustments.

Technical Specifications

Attribute | Value
Manufacturer | Emerson (Epro Brand)
Model Number | CON010 + PR6424/010-040
Sensor Type | Eddy current proximity probe (8mm)
Measurement Range | 0–2 mm displacement
Frequency Response | DC to 10 kHz
Output Signal | 4–20 mA or standardized voltage
Power Supply | 24 VDC (via CON010)
Probe Temperature | –35°C to +180°C
Converter Temperature | 0°C to +70°C
Certifications | ATEX (Hazardous Area), CE, RoHS
Condition | 100% Brand New and Original

Engineering Advantages

Galvanic Noise Rejection: The CON010 converter incorporates full galvanic isolation. This feature eliminates ground loop interference and protects the signal integrity even when the machinery and control room share different ground potentials.

Extreme Thermal Stability: The PR6424 probe tip survives temperatures up to +180°C. This resilience allows for installation deep within turbine housings or near hot bearing pedestals where standard sensors fail.

Simplified Integration: The system transforms complex raw eddy current oscillations into standard 4–20 mA or voltage outputs. This compatibility allows for direct connection to PLCs, DCS, or AMS 6500 racks without additional conditioning hardware.

Precision Machined Hardware: The 40mm sleeve length (designated by the -040 suffix) provides the mechanical rigidity needed to resist high-frequency vibrations within the mounting bracket, ensuring the sensor maintains a consistent gap during operation.

Hazardous Environment Deployment: ATEX certification ensures safe operation in explosive atmospheres, meeting the strict safety protocols of the oil, gas, and petrochemical sectors.

FAQs

Q1: Is the Emerson Epro CON010 + PR6424/010-040 kit brand new?

A1: Yes. We supply this system as a 100% brand new, factory-original unit. Every kit includes original documentation and a 12-month industrial warranty.

Q2: What is the benefit of the integrated 5m cable?

A2: The integrated 5m cable on the PR6424/010-040 reduces the number of connectors between the sensor and the converter. Fewer connections lower the risk of signal degradation and simplify the installation process in oily or humid environments.

Q3: Can this system replace older Epro PR6423 models?

A3: Generally, the PR6424 is the successor to the PR6423 series. While they share similar measurement principles, you must verify the sleeve length and thread type to ensure a proper mechanical fit in your existing mounting bracket.

Q4: Does the CON010 require a specific power source?

A4: The CON010 converter requires a stable 24 VDC power supply. It typically draws power directly from the I/O card or the machinery protection system backplane to which it is connected.